Oil Immersion Objective Question
Posted 11 December 2013 - 09:25 PM
I have the oil to place on a specimen for use with the 1000X.
The images I get, say of bacteria, or cell detail, is clear.
But, there is no more detail seen with the 1000X oil immersion objective than with the 400X non oil objective. Further, the images seen with the 100X objective at 1000X are of equal clarity and detail transmission whether I view with oil or without oil.
Am I doing something wrong? If so, what do I need to change? Or is this the expected behavior of the proper use of an oil immersion objective.
Posted 12 December 2013 - 03:08 PM
That was very helpful.
A few more questions, please.
The 100X objective has the notation stamped on it: 1.25
The oil I am using is labeled 1.5150.
Are these compatible (right for each other)?
Posted 12 December 2013 - 05:34 PM
Your oil objective should look really bad without oil. Is there some oil dried on it? Perhaps you need to clean it. A dirty objective performs poorly too.
Posted 12 December 2013 - 07:07 PM
Is it possible that the condenser needs to be of a certain type? I vaguely remember reading something about a # of .95 not being useful and of 1.25 being useful. I don't know which mine is.
But, the resolution of the 100X is identical to the 400X even when with no oil at all.
Posted 12 December 2013 - 07:25 PM
And the dry and wet oil immersion images have equal resolution with the 400X air objective which provides nice sharp contrasty images.
The scope is a binocular AO Spencer from the 1930s or 1940s I would guess. The microscope is a 40X, 100X, 400X. It did not come with an oil immersion objective and probably was designed only for the 40/100/400. The 100X objective, I picked up from surplus shed.
Posted 13 December 2013 - 02:46 PM
Posted 13 December 2013 - 07:05 PM
I need a little instruction about setting up the specimen for viewing. Let's take blood. I assume I want to smear the blood. Then do I want to cover it with a cover slip or not? If not, do I want to "fix" the smear to the slide somehow so that the oil doesn't loosen and wash it around; say by heating the underside of the slide?
Posted 13 December 2013 - 08:12 PM
The fact you view with and without oil is the same is just odd. I can't get an oil objective to work without the oil. The view is just blurry. Your condenser NA will not affect that. Ideally, your condenser NA should be equal or greater than you objective NA, but you are sill going to use a condenser with a smaller NA, it is just not an an efficient illuminator.
Posted 13 December 2013 - 09:17 PM
Thank you for walking me through this.
I did get a drop of blood and put a cover slip on it. Then I put the oil drop on top of the coverslip and used the 100X objective. The cells seen were larger and the details were more easily seen, then with the 40X objective at 400X. I now understand and see what is meant and what you mean by saying the 100X oil immersion objective provides a better view.
Now, having said that, I want to suggest for your consideration a saying from amateur astronomy. It is said, and it is true, that once something is seen in a finer and larger (aperture) instrument, once the detail is known to exist (say on Mars or the moon), it will then be able to be seen with a somewhat less fine and smaller aperture.
Applying this to my views of the blood. It seems that what I saw at 1000X was bigger, just as contrasty and good resolution, and the detail was clear. But once I knew the detail to look for, I could always find the same at 400X, though it took some active peering. Would you agree, or not agree with this assessment?
And now a question about the blood I was looking at (mine). There were all of the nice round cells. I believe those are red blood cells. There were also cells of the same size as the red blood cells but with all sorts of small bumps and projections on them. There were quite a few of these as well; almost as many as the perfectly round red blood cells. Are these bumpy-cells also red blood cells but somehow deteriorated or degenerated from some gases or lack of gases (oxygen?) in the solution?
Posted 14 December 2013 - 12:33 PM
Also, I run a microscope lab for researchers and my background is in imaging, not biology. So, if there the cells are diseased or they are something else, I probably could not give you an answer.
Posted 29 December 2013 - 07:54 PM
I've done some slides where they show up myself.
Posted 11 January 2014 - 01:38 PM