- My experience using two 80-millimeter long-focus refractors
- GSO 8-inch TRUE CASSEGRAIN
- Celestron Regal 65ED M2
- Review: The Vixen FL55ss
- PrimaLuceLab Eagle Review
- interstellarum Deep Sky Guide Desk Edition
- Chronicling the Golden Age of Astronomy: A History of Visual Observing from...
- Omegon Mini Track LX2 Review
- Review of the APM 152 ED serial number 245
- THE BURGESS 24MM MODIFIED ERFLE & 10MM ULTRAMONO
- APM 140mm DOUBLET APO REFRACTOR
- Comparison of the Boltwood II and Sky Alert Cloud Sensors
- Chile Dilly!
- MONO & BINO VIEWING WITH THE BAADER MORPHEUS 17.5MM EYEPIECE
- The Eye of the Flak (Das Auge der Flak)
CNers have asked about a donation box for Cloudy Nights over the years, so here you go. Donation is not required by any means, so please enjoy your stay.
Burgess Optical 25x100 Binoculars
Discuss this article in our forums
visually observing from East TN since the summer of 2008, with both
binoculars and telescopes. I purchased a pair of Apogee 25x100s a
little over a year ago, and decided to purchase another pair of
25x100s for a variety of reasons. I found a good price on a Burgess
Optical 25x100 here on the CN classifieds, so I picked them
First off, the Apogee and Burgess binos are the exact same body style and armoring. They appear to have been made in the same factory. The main observable differences are:
1. The coatings appear thicker/more colorful on the Apogee objectives.
2. The Burgess pair uses BK7 prisms, light cutoff is clearly visible in the diamond/45° turned square pattern in both eyepieces.
3. The Burgess pair feels heavier (I don't have an adequate scale to measure this), but there is no obvious reason - the externals and internals look the same.
The case that the Burgess pair came with is vinyl/pleather covered with a combo zipper/velcro closure and has some padding and stiffness. You wouldn't want to drop them while they're in this case, but it's good for transporting them for short distances and it has a good shoulder strap attached to it. It is inferior to the Apogee case (nylon/cloth) due to the inability to zipper down the sides – this makes it a nightmare to insert and remove the binoculars.
The objective lens covers fit perfectly.
Now, for the lenses & prisms. The objective lenses appear to be lightly multi-coated on both sides. They are a light green/purple color. Same as the Apogee pair, the eyepieces appear to be only lightly single coated to a light shade of blue. The prisms are definitely BK7 due to the visible cutoff, and it appears that they internal surfaces are lightly multicoated.
The eyepieces are individual focus and are just the right stiffness and have a pretty wide range of adjustment in my opinion, same as the Apogees.
The stiffness of the tripod brace rod in the center is the same as the Apogee pair and it is nice and stiff. The stiffness of the IPD adjustment is very good as well - it doesn't move at all between viewing sessions and transporting it back and forth. The tightening mechanism for the tripod brace rod to tripod head mount is pretty hefty, but you can't get a good enough amount of tightness to prevent a little (not free, you have to push a little) rocking of the binos on the brace rod. It's not a problem, though.
The collimation screws are inside of the binos, same as the Apogees - you have to take the gold colored eyepiece covers off (2 screws) to get at them, and there are 3 per side, but some of them appear to have been cranked on pretty hard as there is obvious usage and what appears to be varnish or some sort of glue to hold them in place. I've got pictures of the collimation screw locations for the Apogees (same for these) in my review of the Apogee pair here:
Now, on to visual
The stated FOV for these binos, printed on the right eyepiece cover, is 157' @ 1000yds (or 3°). The measured FOV, using the Apogee pair for comparison is exactly the same is as the Apogees - 2.25°.
Images are sharp and clear in the middle of the FOV, and are clear & sharp out to about 70%, then drops off sharply to the edge (same with the Apogees). There is the same amount of false color present on these binos as there is on the Apogees. Maximum sharpness & clarity are centered in the center of the FOV.
I've also measured magnification and aperture of these binos using two homemade masks - 80mm & 70mm - over one objective of the binoculars. Measured with a pair of inside calipers, the results are:
100mm - 4.1mm exit pupil = 24.4x
80mm - 3.4-3.5mm exit pupil = 23.2x
70mm - 3.1mm exit pupil = 22.6x
So, it appears that the 100mm is masked down internally. Averaging the 2 closest readings, I get a magnification of 22.9x, and multiplying that by the original 4.1mm exit pupil, I get an actual aperture of 93.9mm. These measurements were with the eyepiece cup off so that the exit pupil measurement was taken right up against the glass of the eyepiece.
IPD measured at 57-75mm, length is 15.25", and eye relief is ~10mm (4mm shorter than the Apogees - the best views were with the eyepieces cups folded away).
Now, for visual comparisons.
Saturn - rings observable in both, but the Burgess pair exhibits some flaring on the left side
M81/82 - slightly brighter in the Burgess pair (compared at least 5 times)
M13 - same in both
M92 - same in both
M65/66 - same in both
M57 - slightly brighter in the Apogees (compared at least 5 times)
One item of note - after a side by side comparison, daytime views of the sky through the Apogee pair seems a little yellow, whereas the view through the Burgess pair seems a little blue. I've attached pics below to show this (the first two pics are courtesy of the previous owner, Jim Rosenstock, used with his permission – thanks, Jim). The pics of the tower (about 3800' away) were taken afocal with a Kodak Z760.
Side by side 1:
Side by side 2:
Burgess view, slightly blue:
Apogee view, slightly yellow: