Cheshire eyepiece to collimate a SCT?
Posted 31 July 2014 - 10:35 AM
Decided to try it with my SCTs.
My 8SE showed perfect concentric circles all centred on the crosshairs. However my 6SE did not- the circles only partly overlapped and were off centre.
No the stange thing is that both SCTs star-test well. Actually the 8SE is slightly better than the 6SE, but there is not much in it.
Should I leave the 6SE as it is or use the Cheshire to try and get "perfect" alignment
Posted 31 July 2014 - 11:08 AM
Perhaps, though you're seeing a decentered secondary? If you collimate by star testing, this will manifest, I believe, as concentric circles on one side of focus, non-concentric on the other, i.e, you'll only be able to get perfect collimation at a specific location of the primary, every other location will show decollimation of the diffraction rings. If this is the case, it's usually sufficient to loosen the corrector retaining ring with the ota pointed up, remove the shims if present, and center the secondary housing by moving the entire corrector, and measuring relative to the corrector cell. Using dial calipers yields precise results. It's less likely, but possible that the secondary housing is off center in the corrector plate, but it's a bit more involved to fix this because Celestron glues the housing in place. Consequently, you hope to be able to accomplish what you need by moving the entire corrector.
Posted 31 July 2014 - 06:33 PM
Posted 01 August 2014 - 07:51 AM
Posted 04 August 2014 - 04:59 PM
Although I have collimated my 8-inch SCT using only a "star test" for concentric diffraction rings in the center of the field of view, I noted that a Celestron instruction sheet/manual (on pg 2) mentions using the Collimation Eyepiece ("Chesire") for collimating an SCT:
I would have never thought to even consider this tool.
Posted 05 August 2014 - 08:45 AM
That's an old document, and as Uncle Rod said above, save it for your Newtonian. The Chesire eyepiece is only effective for rough collimation of an SCT. Same as using the big out of focus doughnut (diffraction rings). For a final accurate collimation you need to to use high magnification and only work just inside and outside of your focus point. Ideally you want to finish using just the Airy disk.